Dynamic Fluid Analysis technology enables Echo Liquid Handlers to adapt easily to different types of fluids. This proprietary technology enables Echo Liquid Handlers to accurately and precisely transfer most reagents, even if the Echo Liquid Handler has never previously transferred that reagent. The result of the Dynamic Fluid Analysis process is that the operator does not have to stop their work and recalibrate, ever.
By removing the calibration requirement, less time is required to maintain and operate a liquid handler. The Echo Liquid Handler uses Dynamic Fluid Analysis capabilities to simplify experimental setup and enables a higher degree of experimental and workflow flexibility.
Echo Liquid Handlers use sound energy to transfer reagents in drop increments on a 2.5 or 25 nL scale (instrument dependent). All Echo Liquid Handlers incorporate a transducer, which converts electrical energy to sound energy to accomplish the drop transfer. The transfer process can be explained in two broad steps:
Key to this process is that the transducer interfaces with one well at a time. This affords the capability to perfect transfer for each source well. General information on the fluid type provided by the operator (e.g., DMSO, buffer, reagent containing glycerol, etc.) narrows the transfer parameters. Dynamic Fluid Analysis uses a power-adjustment ramping process to slowly increase the energy from survey power level to transfer power levels. By listening to subsequent echoes, the Echo Liquid Handler can determine the appropriate power level required for transfer from each source well. This process is accomplished in milliseconds and does not require any operator intervention.